< Previous AED > AED Accessories > Laerdal Accessories > HeartStart FR2 & FR2+ Accessories Philips FR2/FR2+ AVIATION Replacement Long-Life Battery Part #: 989803136291 Our Price: $277.00 Quantity: Previous Next TSO C-142 Designation Aircraft Use TSO C-142 Designation Aircraft Use Call or order online and receive: Same Day Shipping 30-Day Money Back Guarantee 110% Low Price Guarantee Authorized Distributor Exceptional Customer Service Overview Customer Reviews Reference Library Accessories VALUE PACKAGES Overview Philips brand AEDs operate on batteries only. Always have a fully charged spare available. This is the Philips battery pack approved for aviation use in HeartStream/HeartStart FR2 and FR2+ AEDs. It's labeled with the FAA required Technical Service Order (TSO) C-142 designation. These battery packs have an extremely long shelf-life (measured in years) so having an extra one on hand is a great idea. This battery pack can be used on Philips AED models labeled M3840A, M3841A, M3860A, and M3861A only. No reprogramming of the device is necessary to use this battery in an aircraft. Just insert it into your Philips Heartstart FR2 or FR2+ AEDs instead of the regular consumer battery pack and you're good-to-go! This battery comes with a 4-year factory warranty provided it is installed by the shelf-life date stated on the battery. Part 91 Operators: If your aircraft is equipped with a Philips HeartStart FR2 or FR2+ AED, this battery is not mandatory but is recommended. Part 135 Operators: If your aircraft is equipped with a Philips HeartStart FR2 or FR2+ AED, this battery is not mandatory but is highly recommended in the event of an FAA "ramp check." Part 121 Operators: If your aircraft is equipped with a Philips HeartStart FR2 or FR2+ AED, this battery pack is mandatory. Installation procedure The Philips HeartStart FR2/FR2+ Aviation Battery is installed in the same method as any other battery pack for this device. Squeeze the yellow latch on the old battery pack to release and remove it from the AED. Make sure there are no electrode pads connected to the AED. Hold the new Philips Aviation Battery Pack by the yellow latch and carefully slide it in to the battery cavity at the top of the AED. Slide the new Philips Aviation Battery Pack into the AED until you feel and hear it click into place. Once the battery is inserted, your Philips HeartStream/HeartStart FR2/FR2+ will automatically turn on allowing you to initiate a device self-test or perform other functions within the menu system. You'll know that your FR2/FR2+ AED is ready to use when you see the LCD hourglass symbol flashing in the device's self-test indicator window. As usual, always refer to your AED's Operator's Manual for complete instructions. Technical Specifications Battery type: 12 Volt CD, 4.2 Ah, lithium manganese dioxide. Disposable, long-life primary cell. Capacity: When brand new, a minimum of 300 shocks or 12 hours of operating time at 77 degrees F (25 degree C). Shelf life (prior to installation): Usually, 5 years from date of manufacture when stored under standby environmental conditions in original packaging. Standby life (after installation): Usually, 5 years. >4 years when stored under standby environmental conditions (battery installed, FR2 or FR2+ unused). Storage temperature: +32 degrees to 109 degrees F (0 degrees to +43 degrees C). Shipping temperature: -4 degrees to +140 degrees F (-20 degrees to +60 degrees C). Battery limitations: Per battery labeling, never charge, short circuit, puncture, deform, incinerate, heat above 60 degrees C, or expose contents to water. Remove the battery from your AED when discharged. Maintenance and calibration requirements for continued airworthiness: None. For general battery maintenance, please see your FR2/FR2+ Operator's Manual. Aviation Standard Compliance: Meets FAA Technical Standard Order C-142. It is the responsibility of the aircraft owner or crew to install this battery in a specific class of aircraft to determine that the aircraft installation conditions are within the TSO standards. Further documentation is available from an AED Superstore customer service representative. Environmental qualification per RTCA/DO-227, Section 2.3: Passes parameters set forth for vibration, shock, temperature cycle, altitude, decompression, and humidity. Meets following acceptance criteria: No leaking, venting, distortion, fire, or rupture. Change in open circuit voltage <2%. The AED Superstore has a great price on this battery and free ground shipping within the lower 48 United States is included. To place an order just call us and a friendly, knowledgeable customer service agent will help you. Or, click to add it to your cart to purchase online. Read more... Customer Reviews Review this item Reference Library Accessories Related Items Philips FR/FR2/FR2+ Replacement Adult Electrode Pads (2-Pack) Part #: DP2 Our Price: $84.00 ADD TO CART Philips FR2/FR2+ Infant Child Electrode Pads Part #: M3870A Our Price: $110.00 ADD TO CART Philips FR/FR2/FR2+ Replacement Adult Electrode Pads (6-Pack) Part #: DP6 Our Price: $198.00 ADD TO CART The AED Superstore offers the best service, support, and largest selection of AED models for sale available anywhere! We're owned and operated by EMS professionals who continue to use automated external defibrillators in the field. You can count on us to help you make the right decision for your AED program!